AUDIENCE: Biomedical Engineering, Pain Management, Risk Managers
ISSUE: Symbios Medical Products, LLC sent its customers an "URGENT MEDICAL DEVICE RECALL" notification letter detailing the reason for recall and products listed. The reason for the recall is that the flow restrictor bead may become displaced from its fitting which may permit solutions to flow at a higher rate than intended. This product may cause serious adverse health consequences, including death. These kits were distributed between Sept. 10, 2012 and Feb. 11, 2013. Refer to the Recall Notice for a list of kit part numbers.
BACKGROUND: The Symbios Disposable Infusion Pump Kit is a disposable, self-contained infusion system using an inflatable elastomeric reservoir to mechanically provide percutaneous infusion of prescribed solutions at a pre-set rate for post-operative pain management.
RECOMMENDATION: Customers are asked to: segregate recalled product, complete verification form, indicate returned products on verification form, obtain a returned goods authorization (RGA) number, and package the returned products. Refer to the Recall Notice for details.
Healthcare professionals and patients are encouraged to report adverse events or side effects related to the use of these products to the FDA's MedWatch Safety Information and Adverse Event Reporting Program:
[03/18/2013 - Recall Notice - FDA]